Application: Microtype surgical power tools: For application in spinal surgery and plastic surgery. Refined design, the machine is autoclavagle, motor is imported, low noise, high speed, long service life, be warranty for 1 year.
1.Sterilize temperature ≤ 135 °C (Improved type ≤ 155 °C ),if the
temperature is over 135 ° C ,the motor will be damaged.
2.Normal bone drill can only hold drill bits less than 6mm,acetabulum
reaming drill with big torque,rpm is around 300,canulate drill is used
with nail or K wire ,rpm is around 600.
3.The battery need to be get out of the pwer tool after surgery. Otherwise
the small automatic discharge of the electric will make the life of the
4.The battery cannot be sterilized ,it is not autoclavable.
5.The cranial bur must be washed right after the surgery by purger in
order to aviod the blood stick in the bur,then we should make it dry.
6.The battery space in the handpiece of the tool should not have any
7.The cranial bur blade is made of medical stainless steel,in order to
avoid broken,so it is flexible and not very hard,it is easy to be attrite,